Once an extension has been added to the SAAV Extend base array, a new calibration file and site file must be generated to facilitate the collection and conversion of data from the newly combined segments. These files will be built by combining information from the calibration file for the base array and the separate calibration files for any attached extensions.
WARNING: Before connecting to the ShapeArray to generate the required calibration file and site file, you must ensure you have downloaded the latest set of calibration files for the base array and any attached extensions to the computer. This can be accomplished using the Get Calibration Files option in SAASuite while your computer is connected to the Internet.
To generate the new calibration and site files, you must first physically connect a computer running the SAARecorder software to the ShapeArray using one of the following interfaces.
- SAA232-PC cable and an SAA232
- SAA232-PC cable and an SAA232-5
- SAA Field Power Unit (SAAFPU)
- Portable Diagnostics Unit (PDU)
Once connected, start SAARecorder from SAASuite by clicking on the Manual Data Collection button and then the Connect SAA button.
Figure 1. Launch SAARecorder from SAASuite
When the connect screen appears, SAARecorder will attempt to communicate with any connected ShapeArrays. When it detects that an extended ShapeArray is connected and no combined calibration file already exists for that array, it will prompt you to create and save a new calibration file and site file. Click on the Continue button.
Figure 2. New extension detected
By default, a new calibration file for the SAAV Extend base array and any connected extensions will be saved in the C:\Measurand Inc\SAARecorder\Calibrations\Arrays folder. A site calibration file defining the serial numbers of all parts of the complete SAAV Extend array, and the order in which they are physically connected, will also be generated. You must select a location to save these files as they will be required to generate a new logger program to monitor the extended array. The site file created can also be used to rebuild the combined calibration file in the event it is lost.
Click on the Browse button next to the Folder field to select a location to save the site calibration file. Measurand recommends saving this file to the project folder where raw data files from the SAAV Extend were previously saved and converted. Once you have selected the folder and filename for the site file, click on the Save button.
Figure 4. Select a location to save the new combined calibration file and the site file
NOTE: Measurand recommends using the default name for the site file so that it is easily identifiable. By default, this file will have the prefix SAAVExtendsite and will contain the serial number of the base array.
Clicking the Save button will save two new files in the selected location: a site file with the filename specified and a combined calibration file for the extended array.
Figure 5. Combined calibration file and site file saved to the project folder
The combined calibration file’s filename will have the following format.
- SAALift_: The SAALift prefix identifies the calibration file as one for an SAAV Extend ShapeArray with at least one extension installed onto the base array.
- [serial_number]: The serial number of the new top segment of the SAAV Extend after the extension has been added
- [number_segments]: The total number of segments in the extended array
- [segment_length]: The length of the segments in the array measured in millimetres
WARNING: Care must be taken to preserve the combined calibration file and the site file. These files are integral to properly converting the raw data collected from the extended ShapeArray.