Measurand's SAARecorder can run a series of diagnostic tests on a connected ShapeArray and generate a text report with the results. This is helpful for diagnosing the health of the ShapeArray and for troubleshooting any issues with the instrument.
The following steps can only be taken after physically connecting the ShapeArray to the computer and completing the connection with SAARecorder as described in the following step-by-step guides:
- Connecting a ShapeArray to a PC using the SAA232
- Connecting ShapeArrays to a PC using the SAA232-5
- Connecting to a PC using the SAAFPU
- Connecting to a PC using the SAAUSB
Step 1 - Open the Diagnostic Tests Window
From the Diagnostics menu, select the Diagnostics Tests option to open the Choose Diagnostic Test window.
Figure 1: Select the Diagnostics Tests option from the Diagnostics menu in SAARecorder
From the Choose Diagnostic Test window, click the Complete Diagnostic Check button.
Figure 2: SAARecorder's Choose Diagnostic Test window
Step 2 - Choose the Number of Samples for Each Test
You will be prompted to select the number of samples that will be taken for the Voltage / Current, Data Check, and Sensor Noise tests. Enter a numeric value in each of the fields or use the default values by clicking the Use Defaults button.
Figure 3: Define the number of samples to be taken for the Voltage / Current and Data Check tests
Click the OK button to start the diagnostic tests. Multiple windows will open as SAARecorder runs each of the diagnostic tests.
Figure 4: Multiple windows will open and close as the diagnostic tests are executed
Step 3 - Review Results and Save Report
When the diagnostics tests are complete, a Diagnostic Test Results window will present a summary of the test results. You can select the Save text report file checkbox and provide a path and file name for saving a detailed report. The Browse button can be used to select the path and file name. Click the OK button to close this window and save the report. A plain text version and a PDF version of the report will be saved in the specified location.
Figure 5: The results of the complete diagnostics are summarized once the tests are complete